CIP is a concept, in which the cleaning and disinfection dissolution
circulates in the circuit without dismantling the system to be treated.
This is called a circulation, circuit or recirculation cleaning process
that is mainly controlled automatically by programs.
The CIP cleaning concept can be used in systems where circuits can be established, e.g. in tank systems, process lines for the drinking milk production, the butter production, the cheese production, the dried milk production or beverage industry (e.g. in breweries).
The task of the conductivity measuring technology is to control the concentration of the cleaning and disinfection solutions in the supply containers and at decisive points during the process. The concentration in the lye and acid container must kept constant by intensification with concentrated lye or acid. The conductivity of the hot water must be controlled because this is the fluid that comes into contact with the product.
The desalination function is an automatic process, during which the
reference value is reduced by opening a desalination valve after the set
limit value is exceeded.
The water supply that is now started (level control) dilutes the water in the circuit and, thus, reduces its conductivity.
In cooling towers, for example, biocides are added to the cooling water to prevent the growth of biological substances in the cooling systems. The desalination lock is started after the dosing time to ensure that the biocide does not enter the sewage system.
The customer-specific concentration can be used to convert a reference
value to a concentration value by means of a table. Max. 20 support
digits can be used for this. The curve can only be programmed with the
The customer-specific configuration is activated in the analog input conductivity parameter.
Ensure that the 19-digit F No. on the transmitter rating plate complies with the 19-digit F No. on the measuring cell cable end.
If the numbers do not coincide, calibrate the transmitter and the measuring cell using the calibration set (sales article number 20/00459436) (see point 3 in the FAQs).
The language can be changed with the setup program or directly on the device.
From the device software 196.02.01, eight device languages are provided: German, English, Spanish, French, Polish, Swedish, Italian and Portuguese.
Older devices cannot be updated to all current languages.
Changing the language with the setup program:
Changing the language on the device:
The device language is changed in the operator level.